STC 1000 – Digital Temperature Controller
Original STC1000 – Versatile small electronic temperature controller, with heating and cooling functions, plus audible alarm. Always keep your aquarium or Vivarium at a stable temperature of 100% Brand new and high quality. Beer Brewing. Aquariums and Vivarium. This mini Temperature Controller is an excellent and convenient multifunction device for controlling the temperature in Aquariums. The stability of aquarium temperature is one of the crucial factors that contribute to the health and safety of your fish. By utilizing the temperature controller, you can provide an environment with optimal temperature for your fish. This advanced temperature controller can measure and control the temperature from -50 to 90 degrees C. This device is also suitable for Laboratories use and other temperature-controlled systems.
NOTE !!! There are many very much cheaper STC1000’s for sale, beware that these units are not authentic and give many problems
Get the real thing only through us, ask us how to spot the fake ones
STC 1000 – Digital Temperature Controller: Detailed Product Description:
Auto switch between refrigerating and heating
Return difference value
All-purpose Temperature Controller STC-1000:
STC 1000 Features:
STC 1000 can switch modes between cool and heating; Control temperature by setting the temperature setting value and the difference value; Temperature calibration; Refrigerating control output delay protection; Alarm when the temperature exceeds temperature limit or when sensor error.
STC 1000 Specification and Size:
1.Front panel size75(L) -34.5(W)mm;
2.Mounting size:71(L) – 29(W)mm;
3.Product size 75(L)34.5(W) – 85(D)mm;
4.sensor length 2m, including the probe.
STC 1000 Technical Parameters:
1.Temperature measuring range: -50~99oC;
2. Resolution: 0.1oC;
3. Accuracy: 1oC (-50~70oC);
4. Sensor error delay: 1 minute
5. Power supply: 220VAC ±10%, 50/60Hz;
6. Power consumption: 3W;
7. Sensor: NTC sensor (1PC);
8. Relay contact capacity: Cool 10A/250VAC;Heat 10A/250VAC;
9. Ambient temperature: 0~60oC;
10.Storage temperature: -30~75oC;
11.Relative humidity: 20%~85% (No condensate).
Calibrating STC 1000 controller to match thermometer readings. Press and hold “S” display will read “F1” press up arrow, till you see “F4” press and hold “S” you’ll see 0.0 adjust to match thermometer readings (you must keep hold of the “S” while you use up or down button. Make change then press the power button to save the new setting.
How to wire the STC-1000 unit:
In this manual, I will show you how to properly connect and set up a good temperature controller STC1000.
How to properly connect terminals ((Do not connect under electric current))
Please read this verified manual carefully.
STC 1000 manual guide:
Here is a complete easy manual for STC 1000 temperature controller now we describe buttons:
SET button use to set the thermostat, if you Holds for 2 seconds and the thermostat displays the menu with functions F1,F2,F3,F4, (I will explain in detail below)
Arrows are for to increase and lower values
The on-off button is for activation and deactivation
Long push button SET and going in STC 1000 menu F1,F2,F3,F4:
F1: SET GOAL temperature: push SET button again and here see currently set temperature, If you want to change it, press SET again with the up or down arrow (whether or not you want to increase or decrease the temperature). Remember SET + arrow
F2: Hysteresis range : in the manual called “Difference set value ” is a gap when both relays turn off…For example, if you set 1 degree
When the temperature is raised to a target temperature of, for example, 30 degrees of the relay for heating, it is switched off, If the temperature rises further and exceeds 31 degrees, the cooling relay switches on.
F3: Compressor delay time: time for wait after reach goal temperature, if you set 3 min temperature controller wait 3 min, after 3 min turn on the relay.
F4: Temperature calibration: You can make the STC-1000 very accurate, just compare the temperatures with a verified thermometer (best compare at the target temperature) you can increase or decrease the temperature after 0.1C steps.